Level indicator for tank factory China? Ground interference refers to one of the two output ends of the radar level gauge (or compensation line), and its AC voltage to the ground is called the ground interference voltage. This disturbance is also known as longitudinal, in-line mode or in-line state disturbance. Generally speaking, the interference voltage to ground can reach several volts or even more than 100 volts. The above is the interference of the radar level gauge we discussed. When we install or use the radar level gauge, we can also determine whether your use environment will cause these interferences to the level gauge according to the above situation. If these conditions exist, anti-interference measures must be taken to ensure that the level gauge can serve you better. Discover extra information on level indicator.
Any appreciable gain in boiler feedwater achieved through the process reduces the amount of energy (fuel) required at the boiler— in fact, every 10.8°F (6°C) rise in boiler feedwater amounts to a one percent savings in fuel cost. Inadequate level controls can inhibit the deaeration process (level too high) or reduce/shutdown feedwater flow to the boiler (level too low). The former affects hardware longevity and efficiency, while the latter risks production losses and possible damage to pumps.
Secondly, in cement production, material level measurement encountered another difficulty is strong dust interference, especially pneumatic conveying powder silo, dust flying when feeding, low visibility, laser level meter and high energy ultrasonic level meter can not be measured, although radar level meter at this time can receive part of the surface echo. But the echo signal was also weakened. In addition, due to the uneven surface of the radar echo is refracted, will also lead to the existence of radar echo; In addition, there are some conditions from the bottom to the bin inflation, so that the material surface loose, material level measurement is more difficult.
Measurement accuracy, the accuracy of ordinary radar is generally ±10mm, and the accuracy of precision radar is ±3mm. Selection according to the actual needs of production. Range, according to actual needs, choose the antenna size. Note that the actual range is reduced in complex environments. Antenna type and antenna size, the larger the antenna size, the larger the measured range and the stronger the anti-interference ability. The antenna types of radar level gauge are rod type, bell mouth type, paraboloid and so on.
What are the characteristics of radar water level meter? All measuring components are designed in an integrated manner, and there is no mechanical wear during measurement. Because the measurement is a non-contact measurement, it is not affected by the physical properties such as the density and concentration of the water body, it is not easy to be washed away by floods, has a long service life and is easy to maintain. The advantages of the radar level transmitter are obvious. The editor warmly reminds that to choose the right one, it is still necessary to meet your own situation.
KAIDI level transmitter manufacturer is dedicated in providing complete customized solutions for a wide range of industrial automation process applications – in material level, liquid flow, pressure and temperature. We are constantly developing and innovating, our core vision – “to provide solutions that exceed customers’ expectations. In 2012, the company successfully expanded its operations both locally and internationally, achieving global success and recognition for quality fork type level switch, magnetic level gauge products and services. Read even more information at kaidi86.com. Our Radar Level Meter has a range of up to 150 meters, frequency up to 120GHz and an accuracy of ±1mm, which can cope with various complex measurement conditions.
The whole process measurement is rain proof, corrosion proof, high temperature, high pressure, leakage proof, no blind area, eye-catching display, intuitive reading, and a wide measurement range, suitable for all kinds of towers, tanks, tanks, spherical vessels and boilers and other equipment continuous measurement of liquid level. Optional liquid level alarm switch, can realize the upper and lower limit of liquid level alarm and control.
Under many operating conditions, ultrasonic level meter and radar level meter are commonly used. Some users are very entangled in the choice of these two level meter and do not know how to choose. Today, let’s talk about the principles and selection principles of these two types of level meter . Principle and selection principle of ultrasonic level meter. Working principle: The ultrasonic pulse probe emits a beam of ultrasonic pulses to the measured medium, and the sound wave is reflected by the liquid surface. The distance between the liquid levels is measured by measuring the time difference between the emission and reflection of the sound waves. Since the ultrasonic level gauge is not a liquid, it can be used to measure corrosive, non-volatile and non-foaming places.