Top radar level transmitter supplier? The electromagnetic wave emitted during measurement can pass through the vacuum, no transmission medium is required, and the anti-interference ability is strong, and it is not affected by temperature, wind, water vapor, water mist, rain, etc. It can be used for liquid level measurement of almost all liquids, and can be continuously measured on-line unattended. Large measurement range, high precision, mature technology and stable performance. Discover even more information on radar level measurement.
When the distance between the liquid level of the measured medium and the electromagnetic wave transmitter is less than 4m, the guided wave radar is selected. If the distance between the liquid level of the measured medium and the electromagnetic wave transmitter is greater than 4m and less than 35m, select the guided wave radar with the guided wave cable. When the distance between the liquid level of the measuring medium and the electromagnetic wave transmitter is greater than 35m, or when the liquid level of the measuring medium is high temperature, high viscosity or solid liquid level, the air-shooting radar should be selected.
So what can be done about these difficulties? Under the condition of strong dust, on the one hand, the radar with high transmitting energy can be selected, on the other hand, the measurement software with continuous measurement algorithm of wave-loss waiting can be selected. When the radar encounters strong dust, it will not misjudge the measurement result even if the radar loses wave for a short time. After entering the state of continuous measurement algorithm, if the reflection wave of real material surface can be recognized within the set waiting time, the correct measurement value of material surface can be obtained. In the past, only a few foreign radars have this function. At present, there are also domestic radars with this function, and the practical application effect is very good.
Power supply and output signal, the power supply has 220 V A C, 24 VDC, and the two-wire or four-wire system can be selected according to the needs. Output signal 4 ~ 20 mA DC or digital signal, select the appropriate model according to the needs. The above are all factors that need to be considered when selecting a radar level transmitter. The essence of the details is a serious attitude and a scientific spirit. I hope that every user can choose a suitable radar level meter.
The interference comes not only from the outside, but also from the inside of the radar level gauge, such as interference caused by wires, inductance and capacitance between the power transformer and electronic components. In addition, the internal components can also generate noise interference. Today, most radar level gauges have also begun to be improved, using high-frequency microwave technology, which greatly improves the performance of the level gauge and reduces interference.
With emphasis placed on customer satisfaction, innovation, product development and overall business transformation, the company continued to innovate and expand with each passing year. KAIDI has successfully achieved global recognition, obtaining the leading position as Asia’s top process automation sensor manufacturer. In the past 5 years, the company has undergone tremendous growth and development – flourishing internationally and providing customers worldwide with the best customized solutions for process automation. Find more details at https://www.kaidi86.com/. Suitable for chemical industry, petroleum industry, metallurgical industry, water conservancy and electronic industry, etc.
Product features: Large display range, wide range of use, suitable for low temperature, medium temperature, high pressure occasions. It can be matched with remote transmitter output 4-20MADC standard signal to realize remote display of liquid level guage. It can also be matched with liquid level switch to realize liquid level control. Generally speaking, the material level feedback is inaccurate and untimely, which is very prone to explosion events, and there is a greater safety risk. Therefore, the requirements for the radar level meter are very high, but the radar level meter often has abnormal material level jumps or falls, or even crashes, resulting in large errors in material level measurement, which brings trouble to production and sales.